Polyglecaprone 25/ Polypropylene Mesh
“For Better Patient Comfort”
Partially Absorbable Mesh made up of Polypropylene coated with Polyglecaprone 25. Polypropylne mesh is coated with polyglecaprone material.
Partially Absorbable Mesh
Polypropylene mesh coated with polyglecaprone material
Foreign material is reduced to half post complete absorption of polyglecaprone
Burst strenght post absorption is 215N/cm2, which is above maximum intraabdominal pressure
60% reduction in foreign material (post absorption of Polyglecaprone).
Offers higher strength to prevent defect in initial months post operation.
Provides better tissue support for optimum time.
Reduced weight post op offers better patient comfort and formation of flexible tissue scar
Inguinal Hernia: Open & Laparoscopic Repair
Ventral Hernia: Open Repair (retromascular placement)
|Chemical composition||Polyglecaprone 25/Polypropylene|
|Sizes||6cm x 11cm, 7.6cm x 15cm, 10cm x 15cm, 15cm x 15cm, 30cm x 30cm|
|Pore Size||2 mm|
|Polypropylene content||≤ 30 gsm|
|Burst strength - initial||313.8N/cm|
|Burst strength - after Polyglecaprone 25 absorption||215.7N/cm|
|Elongation (% at 1.6kgf/cm in burst strength test)||30 ± 5 %|
|Type of absorption||Partially absorbable|